Industrial endoscope guide inflexible how to solve?
In the inspection process of industrial endoscopes, the implementation process is the result of several functional parts working together, among them, imaging clarity determines whether defects can be found, and the guidance function affects the speed of defect finding. Probe guidance is critical to inspection efficiency, but in use sometimes feel that the probe guidance becomes insensitive and inflexible, this article introduces you to a solution to the inflexibility of the industrial video endoscope probe guidance. First of all, let's look at the principle of guidance. The inspector manipulates the guide lever on the industrial endoscope, the front of the probe guide joint can achieve different directions of bending, so as to achieve the probe guide, this long-distance operation is how the "remote control"? The important thing is the guiding wire that connects the handheld to the guiding joint, for example, the tungsten wire used by JEET industrial endoscopes, which has better ductility. The control of the guiding joystick makes a difference in how tight or loose the wire is, thus pulling the guiding serpentine in the appropriate direction. The guiding wire is usually encapsulated in the guiding channel of the probe wire, usually only a few millimeters in diameter, with other cables in between, so the space of the guiding channel is very compact, and there is also a tighter contact between the wires, and a tighter contact between the wires, in order to reduce friction during the guiding operation and make the guiding guide smoother. Here, the smart you may have guessed the reason for the inflexible probe guide. It is true that the use of solid toner and other lubricating powder is to reduce the guiding friction, at first the lubricating powder is uniformly distributed, but over time, long-term use will cause toner accumulation, indirectly leading to the guide angle decay, the situation described in the opening paragraph. In this case, there is no need to panic, eliminate the cause of the problem can be, jitter probe is an effective solution, can make the lubricating powder redistribution, guide performance and then restore. The specific instructions for shaking the probe are as follows: Generally, two people need to work together, one holding the bottom of the guiding joint, the other holding the root of the probe and gently swinging it from side to side several times. Please note that do not squeeze the guiding joint, because the guiding joint is relatively soft and artificial force to bend it may reduce the guiding life of the device. I believe you have mastered the solution to the inflexibility of the industrial endoscope guide head. Generally speaking, in the daily equipment inspection, and in the inspection process, if a similar problem is found during the inspection process, you can try to solve the problem with the solution of jittering the probe. If there is no improvement, we recommend that you check the probe line as soon as possible to see if there is extrusion deformation or damage, and contact JEET industrial endoscope for maintenance.